A. Multi-Product Calibrator. Service Manual. PN August Rev.5 , The Fluke Model A Multi-Product Calibrator (Figure ) is a precise. Fluke Corporation. 1. A Specifications. The following paragraphs detail specifications for the A Calibrator. changes more than 5 °C. (See “Zeroing the Calibrator” in Chapter 4 of the A Operator Manual.) Also see additional. 2 Fluke Corporation A/A Multi-Product Calibrators Extended Specifications. Warmup Time. Twice the time since last warmed up, to a maximum of
|Published (Last):||8 September 2018|
|PDF File Size:||15.87 Mb|
|ePub File Size:||20.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
calibratr Now adjust R57 until this first ledge is on the horizontal center line. If Fluke determines that the failure was caused by misuse, alteration, accident or abnormal condition of operation or handling. How the Calibration Procedure Works Access procedures are provided for those who want to replace a fluke 5500a calibrator manual module.
Pulse Width Verification The following equipment is used to verify the pulse width.
FLUKE a Calibrator repair help with IFC fet parts & data sheet – Page 1
Equivalent models can be substituted if necessary. Figure shows 550a0 arrangement of assemblies inside the A. Equipment Setup fluke 5500a calibrator manual High Frequency Flatness All high frequency flatness procedures use the following equipment: Operating power for the Current assembly is filtered by the Filter assembly A Edge frequency, duty cycle, rise time Procedure provided In this manual.
Adjust the dc offset on the 1 B so the last ns of the peak of the fluke 5500a calibrator manual wave is on the center line. Perform the “gain” CAL as follows: Signals flu,e 55 V and 95 V p-p are not available. Introduction The Fluke Model A Multi-Produet Calibrator Figure is a preeise instrument that ealibrates a wide variety of eleetrieal measuring instruments. High Frequency Flatness Verification at 25 mV.
Enter text from picture: If this warning occurs recheck the setup and calculation and carefully re-enter the corrected rms reading insuring proper multiplier i. The A prompts you to remove all cables from the front panel outputs.
You may find that you can place the second harmonic at dBc but the third harmonic is less than dBc. SC option must be selected for at least five minutes prior fluuke any capacitance measurement, including the zero process.
Fluke 5500A Calibrator – No Options
Enter into the 5 5 00 A the measured values of each step listed in Table when prompted to do so. These specifications are valid after allowing a warm-up period of 30 minutes, or twice the time the A has been turned off.
Page – Shortcuts for Setting the Frequency and General Specifications Warmup Time Twice the time since iast warmed up, to a maximum of fouke minutes. For the volts-current phase, measure the phase across a noninductive resistor as shown in Fluke 5500a calibrator manual Return to Rl and flatten the ledge as much as possible.
Digital Multimeters (DMMs) and Precision DC Sources
If there are faults associated only with fluke 5500a calibrator manual measurement, the A50 board most likely needs replacement. The delay after the trigger event is also changed for the Edge and Wave Generator functions.
Manually range lock the HP 8 A to the V dc range. This is determined directly in the low frequency band. Harmonics Page – Leveled Sinewave Verification: The supplies are fused on the motherboard. How to Contact Fluke To contact Fluke, call one of the following telephone numbers: Determine that the SC falls within the typical specifications shown in Table To use the fluke 5500a calibrator manual interface, and without having to write a calibration program, do the following: Oscilloscope Input Capacitance Measurement Specifications The signal path is also split to drive the external trigger circuitry on the A50 hoard.
A Operator Manual
Adjust the dc offset on the 1 B so the peak of the square wave is on the center line. Wave Generator Verification at 1 MD Voltage Eunction Specifications Ensure the Fluke 5500a calibrator manual A reading is 0. Once you have turned R44 to the point at which the spur fluke 5500a calibrator manual callbrator, the signal is balanced between the VCOs and you have completed the adjustment.
Then follow these steps to calibrate ac voltage: Remove the two front handles by removing the six Allen screws from the handles. Table shows the steps in this procedure.
See Figure for connections, and see Table for shunt information. Measure the frequency with a precision counter.